Some text in the Modal......
Erecaid Esteem Manual Impotence Pump is the first ergonomically designed vacuum device. Its angled shape fits comfortably in the hand for easy one-hand operation of the pumping mechanism. Restores sexual function in 9 out of 10 impotent men.
Click Here For Detailed Part Information
Start with the insert in the smaller end of the cylinder, as packaged. The insert narrows the inside of the cylinder and keeps scrotal tissue from being drawn in while you create your erection. If your erection presses uncomfortably against the inside of the insert, remove the insert and use the cylinder without it.
Attach the oval end of the pump to the large oval end of the cylinder by pressing them firmly together. Be sure to keep the cylinder’s label face up and opposite to the Pump Handle. Now your ErecAid Esteem System is fully assembled, and you are ready to use the System.
Apply Osbon Personal Lubricant to the following:
It is recommended that you stand for this step (the System can also be used when you are sitting or lying down).
Place the lubricated penis inside the cylinder with the label on the cylinder facing up. With one hand, hold the cylinder at a downward 45° angle with the open end snugly against the body.
Slide the tension rings from the cylinder to the base of your erect penis, one at a time. Keep the cylinder snug to the body while sliding off the rings.
You may also push the green vacuum release button to remove the cylinder. The penis will remain erect until you remove the tension ring(s). Do not leave the tension ring(s) on your penis longer than 30 minutes.
Please enter your and your friends email address below.
Fields marked with * are compulsory